008A CIRS Motion Control Software - June 2016
Attention Users: The 008A CIRS Motion Control Software has been updated to V2.3.0 or V2.4.2, depending on the Serial Number of your controller. Please refer to the 008A user guide for download instructions, or contact CIRS.
Release notes: V2.4.2 (07 June 2016)
- Improvements to the scaling algorithm for imported waveforms
- Added capability to recall previously imported waveforms with the original parameters
- Added the option to gate a DC output signal (valid only for motion controllers with Serial Numbers containing "P821", "P822", or "P824")
- Minor GUI improvements.
ISO Analyze Software - September 2015
Attention Users: The CIRS ISO Analyze Software for use with ISO Cube (Model 023) has been updated to V1.1.2. Please contact CIRS for details.
Version 1.1.2 fixes an algorithm glitch which prevented the collimator ISO Center computations when images of sequences #2 and #3 were acquired with an SID larger than 1600mm.