With CIRS now part of Sun Nuclear, we are collaborating to provide optimal Quality Management solutions for Diagnostic Imaging and Radiation Therapy.
The ACR Digital Mammography Phantom (Model 086), is no longer being manufactured or sold by CIRS. We are pleased to offer the following suitable alternative from Sun Nuclear.
Properly QA your digital mammography systems, with the Mammo FFDM Phantom.
Gain Efficiencies and Image Quality
Perform image quality, artifact detection, uniformity and Contrast to Noise Ratio (CNR) tests using the same window width (WW) and window level (WL).
The Mammo FFDM Phantom simulates radiographic characteristics of compressed breast tissue, including micro-calcifications, ductal fibrous structures and tumor-like masses. Identification of these small structures is essential to the early detection of breast cancer.
The Mammo FFDM Phantom permits testing of the MQSA 3.0 mGy dose limit. Attenuation is equalized inside and outside the wax insert. The test objects are designed and located per the ACR specifications, including the nylon polyamide fibers, spherical glass specks and the spherical cap masses.
- Evaluate artifacts over the entire detector with a single image
- Supports ACR with test objects designed and located per ACR specifications, and reduced backscatter and equalized attenuation
- Meets ACR 2018 Digital Mammography Quality Control Manual requirements
- Supports MQSA 3.0 mGy dose limit testing with attenuation that’s equalized inside and outside the wax insert
1 ACR Mammography Accreditation Program Testing Instructions (https://acredit.acr.org)
Product manufactured by Sun Nuclear; CIRS is now part of Sun Nuclear, a Mirion Medical company.